Fluke 199 Scopemeter
Fra Open Space Aarhus
|This article is a stub. You can help OSAA by expanding it.|
|Auto-Category||Do not hack borrowed|
|Category||Do not hack|
|Type||200 MHz 2.5GS/s DSO|
It seems channel A is somewhat broken as it cannot display the flat parts of sinewave correctly.
It's as though the low frequency response is broken, perhaps there is an input capacitor somewhere that has been overloaded at some point?
Awesome scope lent to OSAA by Scorpio, take extra special care of it and keep all the parts in the carrying case.